- Announcement about May 10th, 2018 update and your password.
I had complained (due to frustration unfortunately rather harshly) in the Sandbox forum about incorrect voltage readings on my prop scope, and have decided to shelve the unit until some better sofware opgrades come along.
Brad's comment about triggering caused me to think again about the scope, and I'm absolutely amazed that no one else has posted about experiencing the same gross error that I observed. In case it went unnoticed, I will repeat my observations here:
On applying a·2.5·V, 1 KHz positive signal on channel 1; at 5V vertival scale it displays correctly, and switching to 2V scale also, as with the 1 V scale. But switch to 0.5 V scale, the top line indicates just below the 1 volt graticule. And also at the 0.2 V scale. At the 100mV scale it reads just below 0.5 volts. Clearly some bad math is happening here.
I'm running 1.05h beta. If other's units don't exhibit these peculiarities, then could my install have been corrupted somehow?
I'd like to use the scope, but if I can't trust it,· I·can't use it. Does anyone else have similar findings??